pendulum shock testing machine
of pendulum shock testing machine “UDAR-1”
|Shock impact acceleration range produced by the pendulum shock testing machine||200…12 000 m/s2|
|Shock impact acceleration duration||0,2 – 5,3 ms|
|Fluctuations of the peak impact acceleration||± 10 %|
|Weight of the calibrated measuring transducers||max 2 kg|
|Operating temperature range||from 18 ºС up to 28 ºС|
|Frame dimensions L×W×H||900×500×830 mm|
|Shock testing machine weight||not more than 100 kg|
of the pendulum shock testing machine “UDAR-1”
The operating principle of the pendulum shock testing machine is based on the ballistic pendulum method. From structural point of view, the pendulum shock testing machine consists of a frame, suspensions holding the hammer, and an anvil with integrated reference primary transducer.
The amplitude of the produced shock pulse is controlled by the reference acceleration measuring transducer. The value of the shock impact acceleration is regulated by the height of the initial hammer position. Shock-absorbers of various thickness and elasticity degree (to be attached to the reverse side of the anvil) are used to produce shock pulses of different duration.
For the calibration (or verification purposes) it is possible to use comparison method of the reference and calibrated transducer, which is mounted on the anvil at the side, which is opposite to the impact side. One can also use the indirect measurements method: measurement of the time necessary to pass the target with the subsequent recomputation of the velocity.
Components of the calibration system
based on pendulum shock testing machine “UDAR-1”
|Ballistic pendulum||produces shock impact acceleration||1 pc||34 486 USD|
|Reference accelerometer||shock pulse amplitude control||1 pc||4 832 USD|
|Spectrum Analyzer ZET 017-U2||recording and control of the signals received from the primary transducers||1 pc||from 3 933 USD|
|Charge preamplifier ZET 440||charge amplifier used for connection of the reference transducers to FFT spectrum analyzer||2 pcs||from 775 USD|
|Reference target||of a certain length attached to the pendulum used for measurement of the pendulum speed||1 pc||included into delivery scope of the ballistic pendulum|
|Contactless optical RPM sensor ВС 401||recording of the speed of the ballistic pendulum against the reference target||1 pc||282 USD|
|ZETLAB Software||Modal analysis software for calculation and display of the measured characteristics||1 pc||included into delivery scope of FFT Spectrum Analyzer ZET 017-U2|
|PC or laptop||upon request||1 pc||from 1 556 USD|
|Training||Customer’s personnel training||10 % from the total price of the system|
|Equipment verification||in cooperation with a certified institution||20 % from the total price of the system|
Total price: from 46 636 USD
Request a Quotation
for pendulum shock testing machine “UDAR-1”
In this video you can see several tests – please, note, that they are not related to calibration or verification.