pendulum shock testing machine
of pendulum shock testing machine “UDAR-1”
|Shock impact acceleration range produced by the pendulum shock testing machine||200…12 000 m/s2|
|Shock impact acceleration duration||0,2 – 5,3 ms|
|Fluctuations of the peak impact acceleration||± 10 %|
|Weight of the calibrated measuring transducers||max 2 kg|
|Operating temperature range||from 18 ºС up to 28 ºС|
|Frame dimensions L×W×H||900×500×830 mm|
|Weight of the hammer||4.8 kg|
|Weight of the anvil||4.8 kg|
|Shock testing machine weight||95 kg|
of the pendulum shock testing machine “UDAR-1”
The operating principle of the pendulum shock testing machine is based on the ballistic pendulum method. From structural point of view, the pendulum shock testing machine consists of a frame, suspensions holding the hammer, and an anvil with integrated reference primary transducer.
The amplitude of the produced shock pulse is controlled by the reference acceleration measuring transducer. The value of the shock impact acceleration is regulated by the height of the initial hammer position. Shock-absorbers of various thickness and elasticity degree (to be attached to the reverse side of the anvil) are used to produce shock pulses of different duration.
For the calibration (or verification purposes) it is possible to use comparison method of the reference and calibrated transducer, which is mounted on the anvil at the side, which is opposite to the impact side. One can also use the indirect measurements method: measurement of the time necessary to pass the target with the subsequent recomputation of the velocity.
Components of the calibration system
based on pendulum shock testing machine “UDAR-1”
|Ballistic pendulum||produces shock impact acceleration||1 pc||25 596 USD|
|Shock vibration transducer AP1012||shock pulse amplitude control||1 pc||from 1 259 USD|
|Spectrum Analyzer ZET 017-U2||recording and control of the signals received from the primary transducers||1 pc||from 3 933 USD|
|Charge preamplifier ZET 440||charge amplifier used for connection of the reference transducers to FFT spectrum analyzer||2 pcs||from 863 USD|
|Reference target||reference target of a certain length attached to the pendulum used for measurement of the pendulum speed||1 pc||included into delivery scope of the ballistic pendulum|
|Contactless optical RPM sensor ВС 402||recording of the speed of the ballistic pendulum against the reference target||1 pc||420 USD|
|ZETLAB Software||Modal analysis software for calculation and display of the measured characteristics||1 pc||included into delivery scope of FFT Spectrum Analyzer ZET 017-U2|
|PC or laptop||upon request||1 pc||from 1 556 USD|
|Training||Customer’s personnel training||10 % from the total price of the system|
|Equipment verification||in cooperation with a certified institution||from 1 668 USD|
POR (price on request)
Request a Quotation
for pendulum shock testing machine “UDAR-1”
Please note that the personal data provided by you is used exclusively for the purposes of fulfilling our contractual obligations, including, but not limited to: issuing invoices, delivery documents, documentation relating to customs clearance process, etc. We do not provide your personal data to any third party except for the purposes relating to our direct contractual obligations. Upon completion of order processing and expiry of the warranty period for the Products provided by our Company, we do not store or process any of your personal data. Please note, that our Company’s Confidential policy does not imply any advertising or marketing activities with the use of your personal data (including your Company name, address, your E-mail, phone number and other information provided by you with a view to placing an Order and organization of the Delivery).