and their transfer to electrical signals by seismic receivers integrated into seismic stations composition
System design variants
Control systems usually have tree-like structure. Seismic control system operating principle is as follows:
Each observation point is equipped with a seismic recorder (velocimeter or accelerometer). Acceleration signals by three space coordinates as well as power supply signal are transmitted from the seismic receiver to the seismic recorder input channels. Seismic receivers can be external (version 1) as well as integrated into the seismic recorder (versions 2, 3, 4).
Seismic recorder performs integrated digital clocks synchronization in real time mode using GPS or PTP as well as digitization of analog signals and transfer of digitized values at the average level of signals processing via Ethernet, USB or WiFi. This level can be implemented as an industrial PC located at the controlled object (versions 1 and 2). Besides the seismic recorders can be connected directly to the server (versions 3 and 4).
The data – events, states and / or constant data flow (16 kbits/sec from each seismic receiver) is transmitted from the industrial PC or seismic recorder to the server by various communications channels. Seismic recorders offline versions can be equipped with solar panels and transfer the data via GSM/GPRS or WiMAX.
Depending on the connection scheme there are several variants of seismic control system topology:
Seismic recorders data synchronous transfer
When observation points are established, there are normally at least two seismic receivers with sensitive velocimeter and a broad-band acceleration meter. In the case if high operational security level is required, up to 6 acceleration meters can be installed.
For the purpose of increasing the system reliability one can install the industrial PC for preliminary data collection and processing. The processed data are more compact in volume if compared to the source data, hence, the data can be transferred via low-speed communication lines (IEC 60870-5-104, MODBUS RTU, dial-up modems, GSM/GPRS or as GSM/SMS notifications).
There are several schemes of seismic recorders synchronization:
individual— the seismic recorder is synchronized by integrated module and GPS antenna;
group — industrial PC or one of the seismic stations is synchronized by GPS, while the remaining stations are synchronized by PTP via internet.
In order to implement synchronization via PTP there should be established a local network using copper or optical wire.
Installation of seismic receivers and seismic recorders
ZET 048 seismic receivers and seismic recorders are installed in special containments and on a concrete foundation.
ZET 048-C seismic recorders are installed on a solid surface and are fixed with mounting panels. The panels are to be selected in accordance with the data specified in the table below:
For installation in horizontal position one should use the element “Plate 1”.
«Wafer 2» is to be used when it is necessary to have adjustment possibility for horizontal mounting – i.e. azimuth rotation (360°) and angle adjustment (within the limit of ±3°).
For horizontal mounting one should use “Plate 1” together with “Mounting bracket” (U-piece min 12 mm with a welded plate with minimum 6 mm thickness). It is allowed to use U-piece for vertical surface mounting (U-piece min 12 mm) without a plate – in this case the U-piece has to be welded directly to the controlled object.
In the case of mounting on a sloping surface the U-piece edge to be welded to the surface should be cut at an angle allowing horizontal (within the limit of ±3°) positioning of the U-piece.
For the purpose of seismic recorder installation on the controlled objects special cabinets can be used – the cabinet bottom should be detachable for the purpose of the sensor installation.
SCADA-project«Seismic Impact Control System»
To place an order for the System, please, fill the check-list below and send by the E-mail firstname.lastname@example.org