Classical shock profile
Measuring equipment, communications-electronic equipment and other products that are exposed to single and multiple shocks in the course of usage and transportation should undergo tests for shock load resistance.
Shock loads resistance tests can be used to estimate sample product quality as well as its structural strength. The tests are performed by exposing the sample to single or multiple shocks with pre-set duration and peak velocity. Below you can see shock pulses shapes produced by the Software Shaker controller-Classical shock as well as the parameters set in the course of program setting.
Shock signals can be produced with pre- and post-shock pulses or without them. In the figure below T stands for shock duration and A for its amplitude. Duration as well as the amplitude of pre- and post-pulses exceeds the duration of shock pulses in 10 times. The trapezoidal signal has additional parameters such as rise and fall time of the signal (T1 and T2). The sawtooth signal has fall time option (T2).
Setting tests profile
in Shaker controller-Classical shock Program
Shock load test parameters are set in the Menu “Vibration tests’ profile editing” available in the main menu “Testing profile”.
Shaker controller Software Classical shock enables setting of the following parameters:
- Duration of the shock (ms) —duration of the pulse, T.
- Shock amplitude, g — peak vibration acceleration value, A.
- Shock frequency per minute— the number of shocks per minute (1 – 120).
- Total shocks number — total vibration test shock number.
- Rise time (ms) — time required to achieve trapezoidal signal maximum value, T1.
- Fall time (ms) — time required to achieve trapezoidal and sawtooth signal minimum value, T2.
- Voltage at FR measuring (mV) — RMS noise signal value used to measure FR prior to tests.
- FR measuring duration, s – duration of time interval prior to tests during which FR is measured using noise signal.
- Checkbox “Pre-shocks” allows to introduce additional shocks with an amplitude of 10% from the main shocks amplitude for the purpose of required voltage precise measurement.
- «Pre-shocks amplitude, g» — is an amplitude of pre-shocks used for the purpose of required voltage precise measurement (maximum 10% from the main shocks amplitude).
- Checkbox «Pre-post-pulses» enables the presence of preliminary signals for balancing of shaker velocity and displacement.
- Checkbox «Seismic shock»enables seismic shock simulation. Unlike ordinary shock, the seismic one has velocity both in positive and negative directions. Seismic sinusoidal signal shape is shown in the figure below
Seismic sinusoidal shock diagram
Drop down list “Control channel” depicts lists of available channels for vibration tests control (channels measuring velocity in «g» or «m/s2».
“Shock direction” option allows to set modulus sign of acceleration amplitude: up (+) and down (-).